Axis Nano™ operates from a notebook under Windows environment and benefits from the power computer technology.
Axis Nano is the a high precision A scan system able to measure all eye types and integrating the latest IOL calculation formulae.
Transfer and storage is made possible to compatible software thanks to the EMR function.
Axis Nano™ adapts to new practice standards and information transfer requirements EMR compatible
Ultrasound biometry remains the only technology that allows measurements in all eye types including when dense cataract is present.
Accuracy is guaranteed thanks to a high signal-to-noise ratio emitter/receiver and adjusted velocities per segment and eye types.
Additionally, the scleral echo discrimination automatically eliminates optic nerve scans.
Together with the contact technique, immersion biometry is also available to avoid any compression of the anterior chamber.
The IOL calculation function allows comparison between different IOL types and calculation formulae:
|Adjustable gain:||20 to 110 dB|
|Time Gain Control (TGC):||0 to 30 dB|
|Transducer frequency:||11 MHz|
|Tip diameter:||6 mm (0.23”)|
|Electronic resolution:||0.03 mm (0.001”)|
|Depth:||40/80 mm sur 2048 points|
|Contact and immersion techniques compatible|
|Aiming beam:||LED or laser pointer*|
Ultrasound propagation velocity adjustable per segment (anterior chamber, lens, vitreous) and IOL and vitreous material
|Built-in pattern recognition:||dense, phakic, aphakic, PMMA, acrylic and silicone material for pseudo-phakic eye types|
|Automatic calculation of standard deviation and average total length||series of 10 measurements|
|Acquisition modes:||automatic, auto + save, manual|
|Automatic detection of scleral spike|
SRK-T, SRK 2, HOLLADAY, BINKHORST-II, HOFFER-Q, HAIGIS
Post-op refractive calculation:
- Pre-op and Post-op refraction, Pre-op and Post-op keratometry
- 6 different methods for keratometric correction and implant calculation:
History derived, refraction derived, contact lens method, Rosa regression, Shammas regression, Double K/SRK-T (Dr. Aramberri’s formula)
7 values bracketed for desired ametropia for each IOL (IOL increment steps: 0.25D or 0.50D)
Simultaneous display of 4 different IOL calculations
|Power supply:||100-240 Vac
± 10% single phase without earth
|Power:||70 VA max|
|Overall dimensions:||15 cm (L) x 7 cm (l) x 2 cm (H)
5.9" (W) x 2.8" (D) x 0.8" (H)
|Weight:||0,125 kg (0.28 lbs)|
|Processor:||Intel® Atom TM CPU 2600|
|Memory:||1024 MD RAM|
|Hard Drive:||320 Go|
|Operating system:||Windows 7 Starter 32-bit|
|Display:||10,1" (25,7 cm) WSVGA (1024x600)
|Integrated Intel® Graphics Media Accelerator 3600 Series|
|Built-in physician and patient database
Compatible with PC and USB video printers
19-20 Octubre 2018
Manta, Manabí, Ecuador
Bascom Palmer XXXX Inter-American Course in Clinical Ophthalmology,
November 4th - 7th 2018, Florida
Oct. 27th – 30th, 2018,
McCormick Place Chicago, IL USA